Ultrasound Probes: 10MHz B-Scan Probe; new 20MHz B-Scan Probe
Axial Resolution: 10MHz: ≤0.1mm; 20MHz: ≤0.08mm
Lateral Resolution: 10MHz: ≤0.2mm; 20MHz: ≤0.15mm
Scan Depth: 10MHz: 28mm-60mm; 20MHz: 19mm-40mm
Scanning Angle: 53°
Dynamic Loop: 10s/100 images, loop or frame-by-frame playback
Image Saving: 100 images
Gray Scale: 256 Levels
Gain: 1-105dB adjustable
TGC: -20dB to 20dB with dynamic range, manual sectional adjustment (6-section adjustable)
Post-processing: Frame-Averaging; Pseudo-Color Codes; Gamma Correction; Text Annotation & Image Magnification
Ultrasound Probe: 10MHz with Fixation Red Light
AL Biometric Measuring Range: Axial Length (AL): 15mm-40mm;
AL Biometric Measuring Accuracy: No more than ±0.05mm;
Total Gain: 98dB, user’s gain adjustability: 1~60dB;
Measuring Mode: 5 groups (Normal, Aphakic, Special, Cataract and Manual Measurement);
Measuring Method: Immersion/Contact;
IOL calculation: SRK-T, SRK-II, BINK-II, HOLLADAY, HOFFER-Q and HAIGIS;
Support IOL calculation after refractive surgery, two groups of formulas can be calculated contrastively and displayed simultaneously.
Automatic measurement and 10 groups averaging and display standard deviation.
Up to 8 groups of IOL constants can be stored.